transformer, test leads, and input jacks — must be purchased from Keithley Instruments. Standard fuses with For analog filter (Models A/A only) . leaving the factory the A and A meet these specifications. the Series A Reference Manual for additional power derating. Keithley A Pdf User Manuals. View online or download Keithley A User Manual.
|Published (Last):||20 November 2006|
|PDF File Size:||16.43 Mb|
|ePub File Size:||10.24 Mb|
|Price:||Free* [*Free Regsitration Required]|
In the command table, the LXI trigger events can be accessed using lan. When called, this function immediately satisfies the event detector, msnual the SMU to continue through the trigger model. Section 3 Functions and features In this section: Anyway, i just want to simply to program the keithley a to run a certain voltage through the hi and lo cables, and read the output current.
To optimize data access, all nodes can cache data from the node that stores the reading buffer data. The positive and negative polarities of the instrument must be calibrated separately.
When enabled, the reading buffer cache improves access speed to reading buffer data. For example, to access status information for the second element of SMU channel A buffer 1, use the following command: The model interactive sourcemeter instrument reference manual is manuxl complete reference. A rising edge is not detected until all external drives release the line and the line floats high. Each of these indicators is described below.
All other characters are left unchanged. For example, trigger events can be triggered when either a manual trigger or external input trigger is detected. Instrument programming The second method you can use to run simultaneous test scripts is to use groups with a single TSP-Link network. For this example, the System SourceMeter instrument is programmed to sweep voltage to 10 V, limit current 6 mA and limit power 50 mW. In the IP address field, enter When sweeping, the actual source output will vary according to the programmed source value until the current limit is reached.
The test system can be expanded to include up to 32 TSP-Link enabled instruments. TSP command reference scriptVar. For detailed information about this topic, see the Status model on pageon page E-1 section of this manual. Set Compliance Limit LAN concepts and settings Confirming port numbers To view the port number assigned to each remote interface protocol: For directions on navigating the menu, see Menu navigation on page Display indicators not shown The items listed below represent the possible display indicators and their meanings.
Turn On Output Connections for voltage calibration B BS Rev.
This value can 2636x be changed with an empty buffer. SynchronousM input trigger Output characteristics: The data must be zero 0 to clear the bit. Pulses can be generated using the functions listed below.
Use the navigation wheel to scroll through the options.
This allows direct control of the line. The table below displays the attributes that you can use to access the reading buffer statistics.
Substitute “smub” for “smua” in all commands. Functions and features Statistic attributes Use the smuX. The table below displays the attributes that you can use keithlye access the reading buffer data. TSP command reference As an example, to set bit B1 of the measurement event reading overflow summary enable register, set status.
It does check to see if all the pulse dimensions can be achieved, and if they can, assigns the keithleu tag or index to the pulse train. Display SMU source-measure readings and compliance values. For example, to set this attribute to 15 you would send: Press the MENU key.